1. The main and mini antechambers cannot replenish air
Reason: ①The air holes in the glove box leading to the main and mini antechambers may be blocked by plastic film, etc.;
②Whether the solenoid valve is not opened.
Method: ①Confirm the position of the air hole according to the direction of the pipeline, and clear the pipeline;
②Check the wiring and solenoid valve.
2. The main and mini antechambers cannot be pumped
Reasons: ①The air extraction holes in the main and mini antechambers are blocked by plastic film or the cabin door leaks;
②The main and mini antechambers door is not closed tightly, or the sealing ring of the main and mini antechambers is loose.
Method: ①According to the direction of the pipeline, confirm the position of the air hole and dredge the pipeline;
②Close the door and check the size of the sealing ring; ③Check whether the operating air pressure is too low (because the main antechamber air valve is a pneumatic valve).
3. The tank pressure continues to increase during cleaning
Reason: The cleaning valve is not opened, the intake pressure is too high, and the pipeline is not dredged.
Method: ①Ensure that the pipeline is unblocked, preferably not to be bent;
②When cleaning, the relative pressure in the tank will continue to be lower than 0.5 mbar.
4. The main valve of the purification column is defective
Reasons: ①The working air pressure is less than 0.5Mpa; ②The main valve signal feedback is faulty.
Method: ①Replace the working gas cylinder, or check the pressure of the auxiliary gauge of the pressure reducing valve at 0.5Mpa; ②Check the signal switch of the main valve.
5. The loop cannot be opened
Reason: low pressure or high pressure maintenance of the fan inverter.
Method: After the voltage is stable, shut down and restart after 1 minute.
View more quality glove boxes with low price.